In the plane of the CCD, the spectrum
scale, expressed in A/mm (Angstroms by millimeter) and also know
as plate factor
with The CCD being sensitive between 4000 A and
9000 A, the linear extension of the spectrum is (9000-4000)/4752=1,05
mm. If the CCD is a KAF-0400, with pixels 9 microns on a side, computing
the spectrum length in pixels is easy: 1,05/0,009=117 pixels. In
general if
In our example, dl=9.10 One should not distinguish between the value of spectrum sampling, as previously computed, with the spectral resolution. The theoretical value of the latter is given by:
with
With
where The main aberration affecting our system is chromatic coma. As its name implies it depends on wavelength. The dimension of the coma as a function of the wavelength is given by
or, expressed as a linear dimension in the plane of the spectrum:
With chromatic coma, resolution is expressed by
We note that resolution is independant of the grating's grooves number and of the grating to focus distance. With N=4 and l=6500 A, the chromatic coma is 152 A, equivalent to an aigrette's length of 0,032 mm. Resolution is then 43. The only way to improve things is to use a slower telescope, but this is less than ideal. First because reducing the field's size is not a good idea if it is desired to observe many objects at the same time, and also because increasing the focal length also increases star sizes, which in turn implies a loss in resolution (read further on). Figure below shows image spots in the focal plane for 3 wavelengths, computed by ray-tracing in the example's optical configuration.
A typical characteristic of a coma, as is visible in the upper figure, is that 80% of the energy is in fact concentrated in one half of the image spot. Hence the previous formulae give a pessimistic result. They are often replaced by the following formulae that better take into account the energy repartition in the image spot:
With these new assumptions, the chromatic coma is only 76 A and the resolution reaches 86. Images on the upper figure are spoiled by some astigmatism. In our disposition it has a linear dimension of
where a and b are expresssed in radians. If a = -b, astigmatism is null for the wavelength l0 corresponding to emergence angle b. To this end one must give the grating an angle such that incidence and diffraction angles be equal, as is shown in this figure:
The a angle that corrects for astigmatism at wavelength l0 is given in radian by
In a spectrograph astigmatism only lowers spectral resolution by a small amount if the detector's surface is located in the same plane as the tangential focal. However, widening in a direction perpendicular to dispersion may cause a loss in detectivity. In our example, widening of the spectrum at l=0,65 micron induced by astigmatism is 0,024 mm, of the same order of magnitude as chromatic coma. Hence giving an angle to the grating with respect to the optical axis gives little benefit with such assumptions.
The last aberration affecting our system is
field curvature, due to the fact that the spectrum does not have
a planar shape but rather a cylindrical one. Of course, since the
CCD detector is flat, focus cannot be readily achieved for every
wavelength in the spectrum. Because of astigmatism, there are in
fact two focal surfaces, one related to the sagittal image (the
image of a pointlike object is lengthened along the dispersion direction),
the other related to the tangential focal (the image is lengthened
perpendicularly to the dispersion direction), with a curvature radius
of
At the wavelength of 0,65 micron (b=3,7°) and for d=21 mm one finds 0,13 mm, which is not to be neglected. Hence, one should not insist on focusing the order 0 image with such a spectrograph, unless the CCD camera is given a slight angle with respect the optical axis so the detector's plane intercepts at the same time the best focus position for order 0 and a given point of the spectrum. It is possible to greatly improve the quality
of our spectrograph by adding an optical component: a small angle
prism with one side positioned against the grating. If the latter
is directly engraved on one face of the prism, then we have an optical
element called a G
If
With our system and for n=1,5 one finds g =4,3°. The length of the residual coma's aigrette for another spectrum point at wavelength l is
The spectrograph's resolution is then in the order of 30A on the full spectrum (see following figure).
Finally, it is very important to remember that the FWHM (Full Width at Half Maximum) directly determines spectral resolution in a slitless spectrograph. Tracking quality shall be maximized, also the telescope should be properly focused and tuned for optical quality. Turbulence itself should be low. For all these reasons, it is often easier to obtain good spectra with short focal lengths than with longer ones (a simple zoom lens corrected for chromatism can be as efficient as a 200 mm telescope for spectral resolution). If the figure for FWHM is 2 pixels and dispersion is 47A/pixel, then it will not be possible to reach a better spectral resolution than 2 x 47 or 94 A. Using numerical applications as described in this page, taking into account the dimensions of your CCD detector, with the goal of studying weak objects' spectra, you should find that the best grating to be used will have at most 70 to 200 grooves per millimeter! 3. AN OPTIMAL GRISM ARRANGEMENT Definitively, when placing the grating very close to the CCD, image quality feels. Figure below shows the spots diagrams for several wavelength when there engraved surface of the Rainbow grating is placed 8 mm in front of the surface of the CCD (dispersion of 55 A/pixel). The analysis of this images shows a chromatic coma which one can reduce by introducing the wedge prism of small angle into the beam, just in front of grating. This prism can be integrated easily in the barrel of the Rainbow grating if it has a diameter of 25 mm.
With a prism of angle 3.87° (H43651 reference
in the Edmund Scientific catalog, material: BK7, thickness: 3.0
mm) one obtains the spots next images. Goto index page Previous page Next page |